General Introduction for W-600 Human / Veterinary Urine Analyzer
W-600 urine analyzer is anather high performance, multi-function medical inspection instruments afterW-200 series urine analyzer. Adopting the advanced “High luminosity cold light source reflection determination” technology and D pipeline architecture, the users put the test strips on the platform at any time and does not need to consider the reaction time. The instruments check the strips with the above sensor, and then transmit the strips to start testing, print the testing results automatically. The information exchange between users and instruments are through the touch screen and interactive programs. Many models of urine strips which are produced by our factory can be used on this urine analyzer. It can finish testes on 12 kinds of biochemical components in urine within 6.9 seconds, such as Leucocyte, Nitrite, Urobilinogen, Protein, pH, Blood, Specific Gravity, Ascorbic Acid, Ketone, Bilirubin, Glucose, and Micro Albumin.
Technical Data Reaction Principle
High luminosity cold light source reflection determination Test items Leucocyte, Nitrite, Urobilinogen, Protein, pH, Blood, Specific Gravity, Ascorbic Acid, Ketone, Bilirubin, Glucose, Micro Albumin. Wavelength 400nm,520nm,560nm ,610nm,660nm,940 nm Test speed 300-600, 522 specimens /hour (affected by human factors and software version) Data Memory 1000 patient results Language Chinese, English Environment requirements Temperature 10℃-30℃, optimal temperature 18 ℃-25℃ Relative humidity ≤80% Communication interface RS-232serial port, parallel printing interface Power supply 220 VAC(±15%), 50/60Hz Fuse 250V 2A Power ≤60w Net weight 7.5 kg Dimensions 350mm×320mm×150mm Data communication interface baud rate 1200bps,2400bps,4800bps,9600bps,19200bps,38400bps,57600bps,115200bps(options) Printer internal thermal printer, outer needle printer port Standard RS232C output port connect with computer.
Urine Analyzer adopts the principle of photoelectric color comparison to test the quantity of biochemical component according to the color change caused by getting the urinalysis strips React with the biochemical components in urine. The instrument uses three kinds of monochromatic light to scan the reagent areas one after another, and the scanning system converts the optical signal to electric signal. The reflection rate value of the reagent area can be calculated according to the strength of the electric signal.