100 Set/Sets per Day hematology analyzer price wholesale price.
Packaging & Delivery
hematology analyzer price wholesale price. Packaging Details:
OEM logo or without logo.
Export standard carton.
Delivery Detail: within 15 workdays after recived 100% payment.
hematology analyzer price wholesale price.
Shanghai, Dalian, Shenzhen
blood chemistry analyzer machine CBC-6000 LED screen auto hematology analyzer Clinical hematology analyzer price
LED screen auto hematology analyzer CBC-6000
This instrument is controlled by an inlaid controller which has powerful function and accuracy. It owns advantages like high accuracy, high speed, excellent repeat function, super automatic and complete function. Compared with the similar products, the rate of the performance and the price are more competitive. The instrument counts by electrical impedance method. It will use three kinds of reagents on it. It can proceed auto-diluents on it. It can do 3-part differential on WBC, and test 22 parameters like WBC,RBC,HGB, PLT, MCV, etc. in high accuracy.
The software has powerful function and excellent save function. It can save all the 35000 tested parameters and 3 histograms. It is very simple to operate. The menu is similar with WINDOWS. The software has powerful function to manage the hardware. It can test and find almost all the faults existing in the instrument. And it is very convenient for users to operate. The instrument can be connected with the printer outside and print very nice test report.
Testing principle: electrical resistance for counting and SFT method for HGB
Clinical function: 3-part diff of WBC, 22 parameters
Working mode: DDR Dual Channel testing and independent hemoglobin testing system
Operation method: keyboard and mouse
Testing speed: more than 60 specimens/min, continuous work in 24 hours
Memory size: store all the specimen parameters, at least 100,000 results and three histogram, parameters text output function
Mode of QC: X-B, L-J
Standardization: manual set and system set
Reference value setting: general and user-defined setting
Accuracy value: WBC (109/L) 2.0%
RBC (1012/L) 1.5%
PLT (109/L) 4.0%
MCV (fL) 0.5%
HGB (g/L) 1.5%
Display: large color LCD display, resolution ratio, 800×600
Information editor: input the information of patient, including name, section, inspection applicator, examiner and verify name
Alert tip: alerting system with 25 items and expert diagnostic system
Input/output: RS232, USB, Printing, Keyboard
Printing report: Chinese/English, pattern is optional
Working environment: temperature: 10°C-35°C, humidity: 10%-90%, available in the plateau environment
Power: AC 220V 50HZ
Dimension: 400mm (L) ×400mm (W) ×460mm (H)
Functions and Modules
The volumetric metering board can calculate liquid volume with the help of volumetric
tubing and optical couplers. It is used for ensuring the accuracy of WBC and RBC/PLT count.
The volumetric metering board has two chambers: WBC chamber and RBC/PLT chamber.
Each chamber consists of a volumetric tube and two optical couplers. When the system starts counting, the liquid starts flowing in the volumetric tube. When the liquid is passing through the first optical coupler, the comparator outputs the signal of starting count. After theliquid is passing through the second optical coupler, the comparator outputs the signal ofstopping count. In this way we cancalculatethe time spent on counting process.
Detection principles of WBC, RBC, PLT
The instrument method accurately counts and sizes cells by detecting and measuring changes in electrical resistance when a particle (such as a cell) passes through a gem aperture sensor.
a) Sample is diluted in a conductive liquid, blood cell is badness conductor, but diluent is well conductor, there is difference between them.
b) As each cell goes through the gem aperture sensor, current goes through the two electrode nearby.
c) When cell goes through the aperture, the resistance is increase as cell volume increase.
d) Treat by magnify circuit, the voltage signal is amplificatory and background noise is removed, then can receive the signal to analysis.
e) WBC is analyzed by the test counter and detect circuit, RBC and PLT are analyzed by another test counter and detect circuit. The MPU analyze and calculate the cells, then give the histograms.
f) The count of PLT adopt the advanced liquid, electron and soft system, it’s can settle the repetitive count of the cells. If RBC enter the analysis area, they will have a similar pulses with PLT, it may cause confusion as repetitive count.